Error -200058Identifier in the list entry is too long. Cannot tristate these lines at this time. Hopefully this has been helpful. Error -200041Calibration constants stored in EEPROM produced an invalid value for one or more analog input channels. have a peek here
Error -200262An attempt has been made to configure a trigger without configuring the appropriate sample clock properties or when Sample Timing Type was set to On Demand. Error -200054EEPROM of the device appears to be corrupt. Make sure that the SCXI chassis is powered on, the SCXI cable is properly connected between the chassis communicator and the SCXI module, and that the cabled module specified in the Error -200033DAQmx Write was invoked more than once between two consecutive sample clocks. http://forums.ni.com/t5/Multifunction-DAQ/DAQmx-Error-Device-identifier-is-invalid/td-p/701876
This should solve the problem. Error -200158Requested operation could not be performed because the necessary digital lines could not be reserved by SCXI. For example, E Series devices use lines 0, 1, 2, and 4 on port 0 to communicate with a SCXI module.
Error -200243Hardware clocking error occurred. All channels must be created before the task is verified. Error -200051Input voltage limits exceeded. Error -200186Channels used to make the connection between two endpoints must be reserved for routing.
Reduce the number of channels. Error -200107Module specified in the hardware configuration is not the module found. I am using NI's DI-DAQ software for Windows version 6.9.3 My boards are PCI-6013 cards with party number 188626D-02. additional hints The conflicting properties must satisfy the following constraint: Master Timebase Rate / Master Timebase Divisor = Counter Timebase Rate Error -200134Counter timebase source and counter timebase rate settings are inconsistent with
This error might be the result of an inadvertent deletion of an NI-DAQmx component. Reduce the sample rate or the number of channels. Error -200247Integer was expected but not found in the script. Refer to user documentation for more information.
I am currently trying to assign the device names, but I'm lost again (sucks to be a newbie). click here now Error -200124Lines 4 to 7 of this port are configured for input. Error -200056Channel name not specified in the list entry. The channel calibration is applied in spite of this because the Apply Calibration if Expired property was set to true.
Any insights as to what I'm doing wrong? http://cygnussoft.com/labview-error/labview-error-1000-run-vi.html Error -200261An attempt has been made to use an analog trigger in multiple situations with differing properties. Error -200031Output gain was coerced to the nearest acceptable value, because the original value was too low given the RF Frequency. Error -200187Channel cannot be connected to itself.
If you’re running LabVIEW 2010, it will be called the Measurement & Automation Explorer. Refer to the documentation for proper syntax for connections involving channel ranges. Error -200004Input termination resistor for one or more channels was overloaded. http://cygnussoft.com/labview-error/labview-error-2628.html If the problem continues, contact National Instruments Technical Support.
Make sure your reference clock is connected and that it is within the jitter and voltage level specifications at all times. Ensure that the device is not being reset. Error -200056The connection count of the attached accessory has exceeded the recommended limit.
Change the subset start offset and/or subset length so the subset fits within the waveform, or increase the size of the waveform. Error -200202Relay name is invalid. Contact National Instruments if the device appears to be functioning improperly. Contact National Instruments if the device appears to be functioning improperly.
This allows the driver to automatically select an appropriate value for the other property. You must open the Configuration panel and associate the NI PXIe-5603 with a supported or external digitizer and a supported or external LO. Error -200048Requested property value exceeds device specification limits. this contact form Error -200108Calibration session is already open on this device.
Remove all of the channels currently in the task except the channel that will be used as the analog trigger source, or change the analog trigger source to a terminal. Call DAQmx Switch Set Topology And Reset. 2. Error -200131Analog trigger source must be the first channel in the acquisition or a valid analog trigger terminal. Error -200226Trigger bus to which the device is connected does not have any free trigger lines for the driver to choose.
You must open the Configuration panel and associate the NI PXIe-5603 with a supported or external digitizer and a supported or external LO. The specified gain and AI Minimum and/or AI Maximum would cause the device to exceed the hardware limit. Refer to the documentation for details about settling time constraints. NITW JAY 0 Kudos Message 2 of 5 (5,718 Views) Reply 0 Kudos 回應： DAQmx Error: Device identifier is invalid.
Refer to the documentation for more information about these three properties. All rights reserved. Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase EnglishChinese(China)KoreanJapanese 44 rating: 1.11 out of 5   Error -200220 With Dennis_Knutson Knight of NI 08-14-2013 05:05 PM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed Highlight Print Email to a Friend Report to a Moderator The simplest thing to Error -200068Scanning is not supported by the specified device.
If you are programming with LabVIEW, use the Active Channel property to specify the individual channel. The device will continue to function, but the accuracy of the measurements may be compromised. Error -200248Specified marker position is too close to the end of the last "generate" statement in a "repeat until" loop. This could cause the counter acquisition to stop.
It will then tell you the device specified and the task name. Device Client: delay between device client installation and device DDR in collection 13. This error might be the result of an incorrect installation of NI-DAQmx or a related software package. Click this arrow.