If that's successful, try importing that task into LabVIEW or running an example program. I am getting this error message after e.g. 5 hours of data acquisition and logging. Above three methods are to solve the error -50405, -50808. Contact us about this article Hello, I have problem with my PCI6115 card. have a peek here
This error occurs again after it. Reset the device from the driver (niScope_ResetDeviceorDAQmxResetDevice). This is because when i execute the program with the electronics unplugged, the loop executes very well without any problems. I have no idea where to start from for debugging.
Self-Calibration failed with 200545 error. It doesnt use modularity as well. I disinstalled and reinstalled the software. If you are still getting this error see if you can capture the exact error message. -50405 isnot an error code for"buffer overflow", which makes it sound like there are multiple
My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. The machine I'm trying to run subjects plumbing fixtures to 250 cycles of alternating hot and cold water. All rights reserved. | United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Most Active Hardware Boards : Multifunction DAQ : Error -50405 on USB 6008 Windows Danke mein wöchentlicher (eigenwilliger) Beitrag zur Innovation http://innovation1.wordpress.com/ 13.08.2009, 09:17 Beitrag #6 Y-P ☻ᴥᴥᴥ☻ᴥᴥᴥ☻ Beiträge: 12.654 Registriert seit: Feb 2006 Developer Suite Core -> LabVIEW 2015 Prof. 2006 EN 71083 Deutschland
When occurred this error, You should make a code like a below picture. It's easy! This error is related to the usb transaction. If you have to run the loops in parallel then make sure look at couple of tutorial on synchronisation and multiple loops design paterns within lapview help.
Administrators 274 5,736 posts Version:LabVIEW 2015 Since:1994 Posted October 9, 2008 QUOTE (CArnold @ Oct 8 2008, 10:05 AM) Every now and then I get the following error message and the Prev by Date: Re: String-to-path does not work on cRIO? The user can reconnect the USB Device. If I try to test one of them the test fails and gives the error -10401 occurred at AI Group Config I am running labview 2012 on windows XP with
To verify that your NI USB device is now working, open Measurement& Automation Explorer (MAX) and expand Devices and Interfaces. Task Name: _unnamedTask<3> Status Code: -50405 at nNIMSSAIL100.StatusObserverT
I would like to also have a digital output signal sent at the same time. navigate here If your task uses a start trigger, make sure that your start trigger is configured correctly. I want to calculate the maximum deviation between rising edge and the falling edge of output signal. Eliminate intermediary connections between the host and the device.
Share this post Link to post Share on other sites Minh Pham 9 Very Active Members 9 171 posts Location:Sydney Version:LabVIEW 2014 Since:2006 Posted October 11, 2008 QUOTE (CArnold @ The machine has four hot solonoid valves, four cold solonoid valves, two zone valves for directing water back to either a hot sump or cold sump, a hot water heater maintained Suche Forenstartseite LVF - NEWS -- LabVIEW Forum News ---- NewsArchiv ---- Systemarbeiten -- NI-News ---- NI-Archive ---- NI-WebCasts -- Forum Feedback & Support ---- Lob & Kritik ---- Tutorials ------ http://cygnussoft.com/labview-error/labview-error-code-9.html mein wöchentlicher (eigenwilliger) Beitrag zur Innovation http://innovation1.wordpress.com/ 11.08.2009, 14:52 Beitrag #4 schrotti LVF-Freak Beiträge: 842 Registriert seit: Feb 2008 2009 - 2011 2006 kA 70180 Deutschland Error -50405 Da kann ich
Every now and then I get the following error message and the machine stops functioning properly: Error 50405 occurred at DAQmx write (Digital ID Bool NChan 1Samp 1 line). My program still runs without any problem but the only thing is that its not writting data into the file after few hours. Can anyone help me regarding this. LabVIEW World 2014년 10월 5일 일요일 [DAQmx] When an error occurred in USB DAQ, the method to restart it programmatically You often find the error -50405 or -50808 when you use
I tried other USB cables and adapters, and nothing worked well except the one included in the package. In some cases your device stops responding and must be reset from the host, either interactively using MAX or programmatically using the driver. Please Contact NI for all product and support inquiries. I dont know how you going to add functionalities nad features in the future.
Also there is a forum discussion that might be useful to you: http://forums.ni.com/t5/Multifunction-DAQ/Problem-with-USB-9233-50405/m-p/959888/highlight/true#M509... Have you tried the device on different USB ports? Recently, all of the channels but one (Channel 3) appear to have stopped generating any signals. http://cygnussoft.com/labview-error/labview-error-code.html Then right-click your NI USB device and click Self Test.
Does this error occur in Measurement & Automation Explorer (MAX) Test Panels? Tried a different port on the computer? Thank you.0 0 07/09/14--12:30: usb 6211 ai GND,AO GND,Ai sense, ai GND problems Contact us about this article Hi all, I have a question about grounding my test shall I use another hardware to control of input data?
Cookies sind für den Betrieb des Forums unverzichtbar. According to Intel, a transaction error is any error that caused the host controller to think that the transfer did not complete successfully. Please tell us why. Related Links: Developer Zone Tutorial: USB Data Acquisition Frequently Asked Questions (FAQ) Community Example:Programmatically Reset USB DAQ device in LabVIEW Attachments: Report Date: 09/05/2007 Last Updated: 05/15/2013 Document ID: 4D4BO9AY
It's a little difficult to see where the porblem is in your code, as you're using lots of express VIs that abstract the lower level functions (including the one with the