If you are resetting the device, wait for the reset to complete. If NI Device Loader still does not respond, check if National Instruments mDNS Responder Service (listed as NI mDNS Responder Service in DAQmx 9.6 or later) is started in Services. Please Contact NI for all product and support inquiries. Call the Traditional NI-DAQ Device Reset VI or the Init_DA_Brds function. have a peek here
Apply Today MATLAB Academy New to MATLAB? Asked by MathWorks Support Team MathWorks Support Team (view profile) 13,642 questions 13,642 answers 13,641 accepted answers Reputation: 2,621 on 27 Jun 2009 Latest activity Edited by MathWorks Support Team MathWorks Apache error AH01797 with trow to white screen server... The NI Device Loader service will automatically start the NI Configuration Manager service.
The device may not be installed on this system, may have been unplugged, or may not be installed correctly. I've tried restarting my system, but to no avail. Find your device by expanding the Data Acquisitionfolder thenuninstall it by right-clicking andselecting Uninstall. (Follow the prompts to uninstall your device.) In Window XP, navigate to your Control Panel and select Also, what version of DAQmx do you have?
This thread profile page shows the thread statistics for: Total Authors, Total Thread Posts, and Thread Activity Home| About Us| Submit Your Site| Update Your Site| Get Search For Your Site| For example, if you have used the device through Traditional NI-DAQ, you must reset the device before the requested route can be made. Error codes with motherboard (no signal Hey i just finished building my pc and went to boot it up but the monitor read no signal. Whenever I run the DAQ assistant configuration, it displays this error.
Solution: This issue is the result of a corrupt DAQmx driver. Neeraj. 0 Kudos Message 1 of 3 (1,210 Views) Reply 0 Kudos Re: Error -89130 occurred at DAQ Assistant P.Sanders Member 12-15-2010 07:56 PM Options Mark as New Bookmark Subscribe Subscribe Show more post info Size: 519 bytes Customize: Reply 2: Re: error 89130 with counters Jim12345678 replied 5 months, 2 weeks ago Tried restarting the PC (strictly a wall powered laptop) Solution: When a computer's operating system (OS) takes a long time to boot, the additional time used by the OS to address the heavy demand of resources from other services may
Possible Reason(s): An internal error occurred.Error -200486 at DAQ Assistant. Opportunities for recent engineering grads. Communication with the device was lost. If National Instruments mDNS Responder Service will not start, please contact NI Support at ni.com/support.
Error. http://digital.ni.com/public.nsf/allkb/19E7993CF8366EED8625764200771C68 Error -89130 occurred at DAQ Assistant Possible Reason(s): Device not available for routing. If the reset fails, try to reinstall the device.First remove the device from the Windows Device Manager. (You can access the Device Manager by right-clicking My Computer and selecting Properties.Next click Showing results for Search instead for Did you mean: Reply Topic Options Subscribe to RSS Feed Mark Topic as New Mark Topic as Read Float this Topic to the Top Bookmark
Import stopped. navigate here United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Most Active Hardware Boards : Multifunction DAQ : DAQmx Error Code -89130: Device is not Available for Routing Poor|Excellent Yes No Document Quality? Call the Traditional NI-DAQ Device Reset VI or the Init_DA_Brds function.
I am using both traditional DAQ and DAQmx on the same system. If not, I would say it is bad, contact NI for repairs. To resolve this issue, either repair the current DAQmx driver that is installed on the system or upgrade to the latest version of DAQmx.These errors can also occur if your DAQmx Check This Out This dependency is removed in DAQmx 9.6 and later.
Reload the page to see its updated state. To reset all devices in Traditional NI-DAQ, right-click the Traditional NI-DAQ Devices folder in MAX and select Reset Driver for Traditional NI-DAQ. Answered Your Question? 1 2 3 4 5 Document needs work? If your device is not detected initially, restart your computer.
Please post your suggestions to get out of this. Error codes i get Ae A0 A6 A3 I have my ssd (intel 480gb) and i dont have my gpu install Error. Import... http://cygnussoft.com/labview-error/labview-error-88705.html Please tell us why.
My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. Related Links: KnowledgeBase 4BHID6DE: Why Are NI Device Loader (nidevldu) and NI Configuration Manager Services Not Starting and Giving an Error -88705KnowledgeBase 7BDCQQNN: Error -89130 When Using Test Panels or DAQ Make sure the service status isStarted. Aftera successful reinstall,conduct aself-test and run some Test Panels in MAX.
For SCXI devices, you must reset the communicator DAQ device. Top contributing authors: Name Posts Jim12345678 2 user's latest post: error 89130 with counters Published (2016-05-05 13:13:00) Tried restarting the PC (strictly a wall powered laptop) etc.. No luck. Took the Click on the Recovery tab and select Restart the Service from the First failure, Second failure, and Subsequent failures listboxes, as shown in the image below. APPCRASH error codes with some programs.
If you are resetting the device, wait for the reset to complete. You have used invalid URL paramaters to request a page.