I booted to save mode, disabled all NI services and then uninstalled NI software which suceeded. One thing was remarkable: The uninstallation failed on the first attempt, it could not stop the services. Restarting LabVIEW or unplugging and re-plugging the device back into the USB port may fix this problem.Error -6006This NI-DAQmx Base error can occur when an incompatible version of NI-DAQmx is installed As i mentioned, setting the recovery options of the service to "restart" doesn't help and a new installation did not help either. Check This Out
I tried this with 9.01, 9.2.2, 9.3. Scott Chan Technical Description of System Error(for Experts only): Microsoft Windows [Version 5.2.4630] (C) Copyright 1985-2014 Microsoft Corp. The device does pass selftest on another machine, it clearly is a software issue and several things point to the problem being the Device Loader service (it is in the state To update the firmware, run the utility FWUpdate from any desktop PC, located here:
If National Instruments mDNS Responder Service will not start, please contact NI Support at ni.com/support. NI Device Loader is dependent on the mDNS Responder Service and will not start if mDNS Responder Service is not working. Yes, after installing the DAQ software the first time i tested it and the selftest passed.
After spending countless hours on this stupid windows labview error 88705 problem I almost gave up. At that point, the workaround will no longer be necessary. If i try to restart it however, it fails to stop, staying in the state "stopping" forever. for instructions to start these two services.2) Computer Restart NeededThis error is sometimes reported if the computer has not been restarted after the installation of the DAQmx driver.
House unusual pop not fact of computer services with to have that to give types of to go the computer system. . ©2014 All Rights Reserved Windows Microsoft Sql Error 7303 Start»Settings»Control Panel»Administrative Tools»Services. The device may not be installed on this system, may have been unplugged, or may not be installed correctly. http://forums.ni.com/t5/PXI/Error-88705-Device-not-found-not-present-PXI-DaqMX/td-p/523465 You can do this by following these steps: Windows 8/7 Log in as an administrator or as a user with administrative privileges.
When I look in Windows Device Manager, my device shows up correctly. STEP 2: Click "Quick Scan" Button to Scan Your Computer. What are some of the common errors that occur with USB devices in DAQmx Base, and how do I fix them? My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation.
This issue should be addressed in an upcoming release of NI-DAQmx. http://digital.ni.com/public.nsf/allkb/B179266E288CE329862573B40078DDC9 NI-DAQmx Base offers a subset of NI-DAQmx functionality on Windows, Linux, Mac OS X, Windows Mobile, and Windows CE OSs. I thought my PC had died when I got this error but now it's as good as new. Luckily, the vast majority of these prospective concerns are unusual.
Did you modifiy something in your computer? his comment is here I.e. The Installation of DAQ-mx 9.4 went fine, all NI services have the state "Started" but the error persists (88705 on selftest). Please Contact NI for all product and support inquiries.
Sadly it didn't change a thing. Is it a problem in the new MAX version?? To start it properly go to Start >> Control Panel >> Administrative Tools >> Services and select the NI Device Loader Service. this contact form Access denied."- Application Log: NI Variable Engine: "...Unable to load the configuration subsystem: Failed to get MAX configuration interface."- Application Log: nipalu, Event ID 12288: "...Could not connect to MXS.
Are there useful logs which could indicate what the cause of the problem is or what next steps could i possibly take? 0 Kudos Message 1 of 34 (2,614 If this does not work I would download a fresh copy of the NI-DAQmx driver, uninstall everything andinstall only that. On the machine we have several other USB-attached Measurement Devices (Agilent, LeCroy and Tektronix), so it is quite possible that one of those is interferring.
This can also occur randomly as an internal software error. My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. Make sure the service status isStarted. Attachments: Report Date: 05/17/2012 Last Updated: 01/23/2013 Document ID: 5XG8M1RP Your Feedback!
no more errors and officially back in action. To correct this problem, you can set these services to automatically restart if they have encountered an error or timeout. All rights reserved. | Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase EnglishChinese(China) 7 rating: 1.85 out of 5   What Errors http://cygnussoft.com/labview-error/labview-error-50202.html By downloading and running the registry repair tool RegCure Pro, you can quickly and effectively fix this problem and prevent others from occuring.
Worked a treat!” Cherelle- 1 Month Ago “Thanks for sharing, I no longer have to put up with the dreaded Labview Error 88705” Sommer- 2 Months Ago “Will this work with To Fix the problem you need to follow the 3 steps : STEP 1: Download & Install RegCure Pro for Free. Error -8002This device initialization error can occur when USB power is temporarily lost from the host, possibly due to operating system hibernation or other power loss. You can find compatible versions of NI-DAQmx in the NI-DAQmx Base readme.
Below are descriptions and potential solutions to some common errors seen when using a USB device with the NI-DAQmx Base driver. Get Your PC Running Normal Again in Under 10 Minutes Filed in Windows Windows Labview Error 88705 on Problem with Windows Labview Error 88705? I cannot tell after what software installation it stopped working, and that is exactly my main issue, i have no idea what interfers. If the error is resolved, you can delete the Data directory that was cut-and-pasted to the desktop.If this does not work I would download a fresh copy of the
If it is not, use the Start button to start the service. Additionally, you may need to run lsdaq and/or close and re-open LabVIEW to reset the software. However, the DAQs are intended to be part of the measurement system, not a standalone thing. Labview Error 88705 is usually caused by a corrupted registry entry.
After the system boots, NI-DAQmx needs some time to figure out what hardware is installed in your chassis and how it is configured before it is ready for VIs to start Unplugging and re-plugging the device back into the USB port may fix this problem. Solution: Important Note:NI-DAQmx Base is different than NI-DAQmx. Error -6004This error typically occurs because the driver associated with the device has recently changed, and the device firmware is incompatible with NI-DAQmx Base.
That's All! Make sure the service status is Started. That's wonderful media since it implies that labview error 88705 odds are excellent that labview error 88705 your difficulty continues to be well documented and can probably be sorted out on So, from my experience, If you received a Windows Labview Error 88705 message then there is a 97.5% chance that your computer has registry problems.
Please Contact NI for all product and support inquiries. United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Most Active Hardware Boards : Multifunction DAQ : Error 88705 with a USB-6366 DAQ Multifunction DAQ Register for See KnowledgeBase 4BHHID6DE: Why Are NI Device Loader (nidevldu) and NI Configuration Manager Services Not Starting?